Outlined below are the electrical specifications for the Dolphin Neurostim™ unit. For additional information, please contact us.
Dolphin Neurostim Safety Report for Pacemaker Patients:
To download a copy of our safety report, please click here.
- Power Source: 9 Volt Battery
- Wave Formation: Square Direct Current (DC wave)
- Voltage Output: 7 – 9 Volts
- Current Output: 50 – 800 microamps (depending on resistance)
(amplitude is inversely proportional to skin resistance V = I x R )
During stimulation phase, parameters are: 0 to 8V with a frequency of 2.5Hz (± 10%). Pulses in this mode also vary with the intensity setting. When set at maximum intensity the current is 60 microamps for 100 kilohms and 400 microamps for 10 kilohms.